Image Map for the vLabtool

Mouse over for details
Amplitude control for the waveform generator Programmable waveform generator output. eg. set_sine(3000.25) Analog input CH1. +/- 16 Volt range Analog input CH2. +/- 16 Volt range Analog input CH3. +/- 3.3 Volt range Manual gain control for CH3. Connect Rg to GND via a resistor to set gain=(1+10e3/Rg). Current to voltage convertor input. read value of 'I2V' CH4-CH7 . 0-3.3 volt , high impedance inputs. SEN . 0-3.3 volt . 1K internal pull-up. Measure Capacitance, knee voltage, resistance etc. 0-3.3 volt. AC coupled inputs for CH1, CH2 Bipolar power supply. 20mA per channel Programmable Voltage Outputs (DACs). e.g set_voltage('PVS1',-1.2) 15KHz fixed sine wave output Square wave generator outputs. Digital inputs Expansion slot Expansion slot for I2C sensors