from __future__ import print_function
from numpy import int16
import time

def connect(route,**args):
	return BMP180(route,**args)

class BMP180:
	ADDRESS = 0x77
	REG_CONTROL = 0xF4
	REG_RESULT = 0xF6
	CMD_TEMP = 0x2E
	CMD_P0 = 0x34
	CMD_P1 = 0x74
	CMD_P2 = 0xB4
	CMD_P3 = 0xF4
	oversampling=0
	NUMPLOTS=3
	PLOTNAMES = ['Temperature','Pressure','Altitude']
	def __init__(self,I2C):
		
		self.I2C = I2C

		self.MB = self.__readInt__(0xBA)

		self.c3 = 160.0 * pow(2,-15) * self.__readInt__(0xAE)
		self.c4 = pow(10,-3) * pow(2,-15) * self.__readUInt__(0xB0)
		self.b1 = pow(160,2) * pow(2,-30) * self.__readInt__(0xB6)
		self.c5 = (pow(2,-15) / 160) * self.__readUInt__(0xB2)
		self.c6 = self.__readUInt__(0xB4)
		self.mc = (pow(2,11) / pow(160,2)) * self.__readInt__(0xBC)
		self.md = self.__readInt__(0xBE) / 160.0
		self.x0 = self.__readInt__(0xAA)
		self.x1 = 160.0 * pow(2,-13) * self.__readInt__(0xAC)
		self.x2 = pow(160,2) * pow(2,-25) * self.__readInt__(0xB8)
		self.y0 = self.c4 * pow(2,15)
		self.y1 = self.c4 * self.c3
		self.y2 = self.c4 * self.b1
		self.p0 = (3791.0 - 8.0) / 1600.0
		self.p1 = 1.0 - 7357.0 * pow(2,-20)
		self.p2 = 3038.0 * 100.0 * pow(2,-36)
		self.T=25
		print ('calib:',self.c3,self.c4,self.b1,self.c5,self.c6,self.mc,self.md,self.x0,self.x1,self.x2,self.y0,self.y1,self.p0,self.p1,self.p2)
		self.params={'setOversampling':[0,1,2,3]}
		self.name='BMP180 Altimeter'
		self.initTemperature()
		self.readTemperature()
		self.initPressure()
		self.baseline=self.readPressure()
		
	def __readInt__(self,addr):
		return int16(self.__readUInt__(addr))

	def __readUInt__(self,addr):
		vals = self.I2C.readBulk(self.ADDRESS,addr,2) 
		v=1.*((vals[0]<<8)|vals[1])
		return v

	def initTemperature(self):
		self.I2C.writeBulk(self.ADDRESS,[self.REG_CONTROL,self.CMD_TEMP])
		time.sleep(0.005)

	def readTemperature(self):
		vals =  self.I2C.readBulk(self.ADDRESS,self.REG_RESULT,2)
		if len(vals)==2:
			T = (vals[0] <<8 ) + vals[1]
			a = self.c5 * (T - self.c6)
			self.T = a + (self.mc / (a + self.md))
			return self.T
		else:
			return False

	def setOversampling(self,num):
		self.oversampling=num

	def initPressure(self):
		os = [0x34,0x74,0xb4,0xf4]
		delays=[0.005,0.008,0.014,0.026]
		self.I2C.writeBulk(self.ADDRESS,[self.REG_CONTROL,os[self.oversampling] ])
		time.sleep(delays[self.oversampling])

	def readPressure(self):
		vals =  self.I2C.readBulk(self.ADDRESS,self.REG_RESULT,3) 
		if len(vals)==3:
			P = 1.*(vals[0] <<8) + vals[1] + (vals[2]/256.0)
			s = self.T - 25.0
			x = (self.x2 * pow(s,2)) + (self.x1 * s) + self.x0
			y = (self.y2 * pow(s,2)) + (self.y1 * s) + self.y0
			z = (P - x) / y
			self.P = (self.p2 * pow(z,2)) + (self.p1 * z) + self.p0
			return self.P
		else:
			return False

	def altitude(self):
		#baseline pressure needs to be provided		
		return (44330.0*(1-pow(self.P/self.baseline,1/5.255)))

	def sealevel(self,P,A):
		'''
		given a calculated pressure and altitude, return the sealevel
		'''
		return(P/pow(1-(A/44330.0),5.255))

	def getRaw(self):
		self.initTemperature()
		self.readTemperature()
		self.initPressure()
		self.readPressure()
		return [self.T,self.P,self.altitude()]
		
